uRADMonitor "Industrial" Multi Sensor Air Quality Monitor
An automated, fixed monitoring station that tracks a total of 12 important air quality parameters including Particulate Matter, Carbon Monoxide, Ozone, Sulphur Dioxide, Nitrogen Dioxide and more. It comes in a rugged aluminum enclosure with wall mounting support. The data is exported to the uRADMonitor network with Wifi or LORAWAN, and can be accessed in real time using the cloud API interface or directly via the local network.
|Sensor||Parameter||Minimum value||Maximum value|
|Bosch BME280||Temperature||-40 °C||+85 °C|
|Pressure||300 hPa||1100 hPa|
|Humidity||0% RH||100% RH|
|Winsen ZH03A||PM1.0||0 μg/m³||1000 μg/m³|
|PM2.5||0 μg/m³||1000 μg/m³|
|PM10||0 μg/m³||1000 μg/m³|
|Winsen MP503||VOC||10 ppm||1000 ppm *|
|* Winsen ZE03-O3||Ozone||0 ppm||20 ppm|
|* Winsen ZE03-CO||Carbon Monoxide||0 ppm||1000 ppm|
|* Winsen ZE03-SO2||Sulphur Dioxide||0 ppm||20 ppm|
|* Winsen ZE03-NO2||Nitrogen Dioxide||0 ppm||20 ppm|
Custom gases detection options* The 4 electrochemical sensors can be replaced with any combination of sensors for the following gases and ranges:
|CO||Carbon monoxide||0 - 1000ppm||0.5ppm|
|O2||Oxigen||0 - 25%VOL|
|NH3||Ammonia gas||0 - 100ppm||0.5ppm|
|H2S||Hydrogen sulfide||0 - 100ppm||0.1ppm|
|SO2||Sulfur dioxide||0 - 20ppm||0.1ppm|
|NO2||Nitrogen dioxide||0 - 20ppm||0.1ppm|
|Cl2||Chlorine gas||0 - 10ppm||0.1ppm|
|O3||Ozone||0 - 20ppm||0.02ppm|
|H2||Hydrogen gas||0 - 1000ppm||2ppm|
|HF||Hydrogen fluoride||0 - 10ppm||0.1ppm|
|C2H4||Ethylene||0 - 100ppm||0.5ppm|
|CH2O||Formaldehyde||0 - 10ppm||0.1ppm|
|ETO||Ethylene oxide||0 - 20ppm||0.1ppm|
|HCl||Hydrogen chloride||0 - 20ppm||0.1ppm|
|C6H6||Benzene||0 - 100ppm||0.1ppm|
|C7H8||Toluene||0 - 500ppm||1ppm|
|C2H3Cl||Vinyl chloride||0 - 20ppm||0.5ppm|
|C2H6S||Methyl Sulfide||0 - 100ppm||0.1ppm|
|C2H6S2||Dimethyl Disulfide||0 - 100ppm||0.1ppm|
|AsH3||Arsine||0 - 3ppm||0.01ppm|
|C3H9N||Trimethylamine||0 - 100ppm||1ppm|
|C8H8||Styrene||0 - 100ppm||0.1ppm|
|CH4S||Methanethiol||0 - 100ppm||0.1ppm|
|CS2||Carbon Disulfide||0 - 100ppm||0.1ppm|
|PH3||Phosphine||0 - 10ppm||0.05ppm|
|HCN||Hydrogen cyanide||0 - 100ppm||0.2ppm|
The National Emissions Ceiling Directive caps some emissions including particulate matter (PM) and nitrogen dioxide (NOx) at national level. A revised version of the directive is as of 2016 under scrutiny by the Council of Ministers and European Parliament. Across the EU in 2013, nitrogen dioxide (NO2), which is mostly produced by traffic, caused 68,000 premature deaths. The Dieselgate scandal exposed how Volkswagen had gamed NO2 emissions tests.
Ozone (O3) killed 16,000 and small particulate matter (PM2.5) caused 436,000 deaths in the same year. PM2.5 particles, microscopic specks of dust and soot caused by burning fossil fuels, can enter the lungs and bloodstream.
Air pollution has different particulate matter (PM) components – smoke, dirt and dust form coarse particles known as PM10 and metals and toxic exhaust from smelting, vehicle exhaust, power plants and refuse burning forming fine particles called PM2.5.
uRADMonitor INDUSTRIAL is equipped with all sensors required to compute the Air Quality Index as defined by several international standards on air quality and give a direct assessment on the pollution problem and possible infringements.
|Voltage||External||6V - 28V|
|Connectivity||4 options||WiFi, LoRaWAN, USB|
If you have a radio variant with an antenna, connect the antenna first. Next, connect the uRADMonitor to the power source using a DC adapter with voltage between 6V and 28V. Connect the uRADMonitor to the power source using a DC adapter with voltage between 6V and 28V or via the USB port with a 5V USB adapter.
The WIFI variant
Use a smartphone or a computer with WLAN capabilities to connect to the local hotspot spawned by your A3 unit. The SSID is uRADMonitor-XX, where XX are the last two digits of the Device ID number. The key is the Device ID, in uppercase, as printed on the enclosure. Open 192.168.4.1 in your browser, and click the "WIFI" link to setup the connection to the Internet AP. Enter the SSID and key of your Internet Access Point. If the connection fails, you will see the status message, and three consecutive beeps will indicated the problem. Alternatively you can change the configuration using the USB data port. See the USB manual.
The LORAWAN variant
Your device must be pre-provisioned with the LoraWAN Gateway and network server details. Alternativelly you can change the configuration using the USB data port. See the USB manual.
The device comes with an USB port, so the configuration settings can be customized via the USB connection. Consult your product manual for complete details.